Fw: [softrock40] Dynamic range measurements of SR V6
wb4jfi at knology.net
Mon Oct 2 11:56:10 CDT 2006
From the Softrock group.
----- Original Message -----
To: softrock40 at yahoogroups.com
Sent: Monday, October 02, 2006 12:38 PM
Subject: [softrock40] Dynamic range measurements of SR V6
I performed some more tests on IMD and dynamic range of the SoftRock V6 (original and modified). Here is a summary of the results.
For all measurements, the reference bandwidth is 2.4kHz, the signal spacing is 15kHz and the frequency is around 7050kHz.
Instuments used: two HP 8640B signal generators, HP 8494A and -B step attenuators, Olektron power combiner. Levels checked with an HP 8566B spectrum analyzer. Power bandwidths and S/N measured with a Fluke 8920A digital RMS voltmeter. All instruments have traceable calibration.
1) Unmodified SoftRock, Soundblaster USB 16bit audio interface (16bit, 48kHz, better antialiasing and response than the usual AC97 found in PCs)
blocking dynamic range: 96dB
IMD3 dynamic range: not measurable (blocking occurs by saturation of the audio interface)
2) same as (1), soundcard Audiophile 192 (24bit, sampling at 96kHz)
Blocking dynamic range: 100dB
IMD3 dynamic range: 93dB
3) same as (2), post-mixer amp voltage gain reduced from 500 times (5kOhm/10Ohm resistors on the opamp) to 90 times (900Ohm/10Ohm)
Blocking dynamic range: 110dB
IMD3 dynamic range: 96dB
4) same as (2), post-mixer amplifier has been replaced by a couple of OP27G powered from +/-12V. Opamp gain is now 200 times (2kOhm/10Ohm)
Blocking dynamic range: 115dB
IMD3 dynamic range: 102dB
a) I don't see any saturation in the FST3253 at least up to +10dBm (not checked over that)
b) the IMD products seem to come mainly from the opamp and audio interface. Low power, low voltage opamps are the limit here
c) some front-end gain is advisable to raise the MDS and stabilize the impedance seen by the mixer, if the opamp and audio interface can handle the strong signals. Noise from the opamp dominates the noise floor, of course; measured noise figure here is approx. 30dB
d) IIP3 figures appear very good, but the real performance is limited by the poor image suppression and the 2nd order audio harmonic generation in the chain. With only two tones at the input the spectrum becomes a real forest, such that it's difficult to find a clean spot to see the IMD products. So, probably we have to talk about SFDR (spurious free dynamic range) or the system. Much harder than a simple IMD3 measurement.
Next step: to test different mixer topologies (switch followed by an active integrator, as proposed on the HPSDR group, and some other ideas).
73 - Marco IK1ODO
Messages in this topic (2) Reply (via web post) | Start a new topic
Messages | Files | Photos
Change settings via the Web (Yahoo! ID required)
Change settings via email: Switch delivery to Daily Digest | Switch format to Traditional
a.. 45New Members
b.. 18New Files
Visit Your Group
a.. Ham radio antenna
b.. Ham radio repeaters
c.. Ham radio sales
d.. Ham radio store
e.. Ham radio outlet
Staying in tonight?
Check Daily Picks &
see what to watch.
Share Your Passion
Join the web's lar- gest community.
Start a group
in 3 easy steps.
Connect with others.
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the Tacos